Search
• Papers

Automated control system cryostatting of receivers RT-32 “Holod”

Transactions of IAA RAS, issue 47, 23–30 (2018)

Keywords: microcryogenic system, cryostat, RT-32, programmable logic controller, control.

Abstract

The article describes the problems of operating the existing micro-cryogenic system. Outlined ways to solve these problems. The purpose of this work is to create a modern, automated, operational, informative and reliable process control system for cryostatting of RT-32 receiving devices. The work on the creation of an Automatized system of cryostatting of receiving devices was carried out in stages. At the first stage, bench tests of the existing control system were carried out. The second stage is the stage of mock tests of the system being developed. The final stage was the trial operation of the system as part of the existing micro-cryogenic system based on the Svetloye Observatory. The introduction of the Automated system for controlling the cryostating of receiving devices increases the autonomy and reliability of the micro cryogenic system, and it becomes possible to visually monitor the cryostat process. The automated control system for cryostatization of receiving devices has been developed and implemented at the Svetloye Observatory, in the future this system will be installed in all observatories of the Kvazar-KVO complex.

Citation

Text
BibTeX
RIS
A. V. Isaenko, I. A. Rahimov, I. A. Pozdnaykov. Automated control system cryostatting of receivers RT-32 “Holod” // Transactions of IAA RAS. — 2018. — Issue 47. — P. 23–30. @article{isaenko2018, abstract = {The article describes the problems of operating the existing micro-cryogenic system. Outlined ways to solve these problems. The purpose of this work is to create a modern, automated, operational, informative and reliable process control system for cryostatting of RT-32 receiving devices. The work on the creation of an Automatized system of cryostatting of receiving devices was carried out in stages. At the first stage, bench tests of the existing control system were carried out. The second stage is the stage of mock tests of the system being developed. The final stage was the trial operation of the system as part of the existing micro-cryogenic system based on the Svetloye Observatory. The introduction of the Automated system for controlling the cryostating of receiving devices increases the autonomy and reliability of the micro cryogenic system, and it becomes possible to visually monitor the cryostat process. The automated control system for cryostatization of receiving devices has been developed and implemented at the Svetloye Observatory, in the future this system will be installed in all observatories of the Kvazar-KVO complex.}, author = {A.~V. Isaenko and I.~A. Rahimov and I.~A. Pozdnaykov}, doi = {10.32876/ApplAstron.47.23-30}, issue = {47}, journal = {Transactions of IAA RAS}, keyword = {microcryogenic system, cryostat, RT-32, programmable logic controller, control}, pages = {23--30}, title = {Automated control system cryostatting of receivers RT-32 “Holod”}, url = {http://iaaras.ru/en/library/paper/1889/}, year = {2018} } TY - JOUR TI - Automated control system cryostatting of receivers RT-32 “Holod” AU - Isaenko, A. V. AU - Rahimov, I. A. AU - Pozdnaykov, I. A. PY - 2018 T2 - Transactions of IAA RAS IS - 47 SP - 23 AB - The article describes the problems of operating the existing micro- cryogenic system. Outlined ways to solve these problems. The purpose of this work is to create a modern, automated, operational, informative and reliable process control system for cryostatting of RT-32 receiving devices. The work on the creation of an Automatized system of cryostatting of receiving devices was carried out in stages. At the first stage, bench tests of the existing control system were carried out. The second stage is the stage of mock tests of the system being developed. The final stage was the trial operation of the system as part of the existing micro-cryogenic system based on the Svetloye Observatory. The introduction of the Automated system for controlling the cryostating of receiving devices increases the autonomy and reliability of the micro cryogenic system, and it becomes possible to visually monitor the cryostat process. The automated control system for cryostatization of receiving devices has been developed and implemented at the Svetloye Observatory, in the future this system will be installed in all observatories of the Kvazar-KVO complex. DO - 10.32876/ApplAstron.47.23-30 UR - http://iaaras.ru/en/library/paper/1889/ ER -